Welcome to LMTmag.com, the social and information hub of the dental lab industry. Connect with industry peers and vendors, ask questions, sign up for events, review products, read LMT articles and industry news and more!
I have been using the same metal for years and the sprues are short and 8 gauge , the burn out is good because the other rings are coming out I only have...See more trouble with the one type of metal and have been using it forever. Anybody have any suggestions?
Well I think I figured it out about the miscast. Thank you for all suggestions, what the problem was not enough oxygen in the gas mixture. Once the oxygen...See more level was moved up the problem was solved. I guess the flame was not hot enough even though I had sharp edges on my buttons.
1). If u r burning out more than 1 ring at a time. Burn out this problem metal by it self you could have a cold spot in the burn out oven. Or have...See more just overloaded the ovens ability .
2) if already burning out by its self increase the burn out temp by 100 degrees F. I have found this to help many...
Demand for flexible partials is on the rise, with about half of respondents to LMT's 2012 Removable Survey selling more flexible partials than they did five years ago. And among those who are fabricating partials as a service to other labs, 21% note an increase in the demand for flexible partials.
84 South Main St., Newtown CT 06470 203-426-4LMT · 203-491-2300 (fax) info@LMTmag.com
LMT Communications delivers the business strategies, information, dental laboratory news and education that dental laboratory decision-makers need to succeed. Through LMT Magazine and our LAB DAY shows, LMT works to advance excellence in dental technology, enhance the products, services and communication of dental technicians, and give lab owners and managers the tools they need to flourish. Learn More »